j. Place POWER switch to ON,
2-14. R197 Adjustment Procedure (fig. 2-3)
Perform adjustment of RI 97 as follows:
High voltage is present. When power is applied to test set, voltages as high
as 115 vac and 300 vdc are present. Contact with high voltages may cause
injury or death.
b. Manually activate interlock switch S9.
c. Reconnect ac and dc POWER cords.
d. Connect DVM (AC mode) between J3-z and J3-q.
e. Place POWER switch to ON.
f. Adjust R197 on circuit board to 3.08 vac 0,05 vat.
g. Place POWER switch to OFF.
h. Disconnect ac and dc POWER cords.
j. Place POWER switch to ON.
a. Remove ac and dc power cords from inner case cover storage compartment.
b. Remove the twelve screws and washer from the front panel edge.
It maybe necessary to carefully pry the edge of the panel to break the seal if
the gasket is stuck.
c. Lift test set from case using handles provided.