DIRECT SUPPORT AND GENERAL SUPPORT
Section I. SERVICE UPON RECEIPT OF MATERIEL
(2) All switches, lamps, and connectors on
applicable test equipment must be intact.
a. When new or reconditioned Laser Test Sets or
Alignment Kit are received by the using organization,
(3) All switches must function normally.
check that all items are present, in good condition,
(4) All panel markings, and other essential
clean, and properly stored. Purge and charge the tester,
markings must be legible.
(5) Cables and extender card connectors and
other accessory equipment must be in good condition.
3-2. Modification Work Orders.
There must be no brittle or frayed insulation.
All pertinent modification work orders must have
(6) The special tools and test equipment must
been applied. Check DA Pamphlet 310-7. If applicable,
be examined for evidence of abuse.
check on the application of all authorized modifications
b. Specific Requirement. Test Sets and Alignment
to see that no unauthorized alteration has been made,
Kit components should conform to the configuration
or that no work beyond the authorized scope of the unit
is being attempted.
The R/T tester and PFN charge ammeter should be in
good operating condition.
3-3. Receiving Inspection.
a. General Requirements.
3-4. Services for Deficiencies on Receipt.
(1) The external case and cover must be
Any deficiencies discovered by the inspection
clean and in good condition.
Section II. REPAIR PARTS, SPECIAL TOOLS, AND EQUIPMENT
3-5. Repair Parts.
3-6. Special Tools and Equipment.
Repair parts, special tools, and equipment for
the test set at direct and general support maintenance
levels are contained in TM 9-4931355-24P.
Table 3-1. Special Tools and Equipment
To purge and charge tester.
Used with Kit, Purging 4931-00-065-